Plastic, gel-filled, digital, double-junction probe with 1.5 meter cable. pH Range: 0.000 to 14.000 (°0.004, accuracy of sensor electronics). Temperature Range: 0 to 80°C (32°F to 176°F). Temperature Accuracy: °0.2°C. Dimensions: 4.7”L x 0.47” dia. Warranty: One year.
Glass, digital, double-junction probe with 1.5 meter cable. ORP Range: -1200 to +1200 mV (°0.2 mV). Temperature Range: 0 to 100°C (32°F to 212°F). Temperature Accuracy: °0.2°C. Dimensions: 4.7”L x 0.47” dia. Warranty: One year.
Epoxy body and 4-electrode, graphite, digital probe with 1.5 meter cable. Conductivity Range: 10 µS/cm to 2,000 mS/cm (°0.5% of value). Resistivity: 0.5 Ohm cm to 100 kOhm cm (°0.5% of value). Salinity: 0.0 to 70.0 ppt (°0.5% of value). Total Dissolved Solids: 0 to 1,999 mg/L; 0.0 to 199.9 g/L (°0.5% of value). Temperature Range: 0 to 100°C (32°F to 212°F). Temperature Accuracy: °0.2°C. Cell Constant: 0.475 cm °1.5%. Dimensions: 4.7”L x 0.6” dia. Warranty: Two years.
Optical sensor with lifetime luminescence detection with self-stirring mechanism. Features a 2m cable. Dissolved Oxygen Range: 0 to 50 mgL; 0 to 500% air saturation. Dissolved Oxygen Accuracy: 0 to 20 mg/L, °0.1 mg/L or °1% of reading, whichever is greater; 20 to 50 mg/L, °10% reading (system). Dissolved Oxygen Resolution: 0.01 mg/L; 0.1%. Temperature Range: Ambient 10°C to 40°C (50°F to 104°F); Compensation mg/L -5°C to 50°C (23°F to 104°F) (extrapolates beyond 45°C). Temperature Accuracy: °0.2°C. Temperature Resolution: 0.1°C. Typical Response Time: 95% in 22 seconds with stirring; 95% in 40 seconds without stirring. Warranty: Two years; One year DO sensor cap.
Automatically calculates BODs/CBODs and improves data integrity by eliminating human error in calculations. Use to create daily bench sheets from established groups and samples. All batches are saved for retrieval at any time. User-configurable setup regardless of seeding method, number of dilutions, or other test criteria. Operating Requirements: Microsoft Windows® XP, Vista, or 7.
Have a problem, comment or any issue with this page? Then we want to hear from you! If you have a suggestion on how we can improve this page, please let us know in the comment space below. Please, provide as much detail as possible to help expedite your request.